|Author||Namboodiri, Narayanan ♦ Shajeem, O. ♦ Tharakan, Jaganmohan Achuthan ♦ Sankarkumar, R. ♦ Titus, T. ♦ Valaparambil, A. K. ♦ Sivasankaran, S. ♦ Krishnamoorthy, K. M. ♦ Harikrishnan, Sivadasan Pillai ♦ Dora, S. K.|
|Source||Sree Chitra Tirunal Institute for Medical Sciences & Technology|
|Publisher||European Journal of Echocardiography|
|Subject Domain (in DDC)||Technology ♦ Medicine & health ♦ Diseases|
|Subject Domain (in MeSH)||Cardiovascular Diseases ♦ Diseases ♦ Diagnosis ♦ Equipment and Supplies ♦ Analytical, Diagnostic and Therapeutic Techniques and Equipment|
|Abstract||Aims TTK Chitra heart valve prosthesis (CHVP), a tilting disc mechanical heart valve of tow cost and proven efficacy, has been in use for the last 15 years. Although various studies substantiating its long-term safety and efficacy are available, no study had assessed its echocardiographic characteristics. The purpose of this study was to determine the normal Doppler parameters of CHVP in the mitrat position and to assess whether derivation of mitrat valve area (MVA) using the continuity equation (CE) and more commonly used pressure half-time (PHT) method is comparable in the functional assessment of this tilting disc mitral prosthesis.Methods and results Doppler echocardiography was performed in 40 consecutive patients with CHVP in the mitrat position. All patients were clinically stable, without evidence of prosthetic valve dysfunction such as significant obstruction or regurgitation, enclocarditis, left ventricular dysfunction (ejection fraction <40%), or significant aortic regurgitation. Valve sizes studied included 25, 27, and 29 mm. Mitrat valve area was derived both by the PHT method and by the CE, using the stroke volume measured in the ventricular outflow tract divided by the time-velocity integral of CHVP jet. The peak Doppler gradient ranged from 5 to 21 (mean 11.0) mmHg, and the mean gradient ranged from 1.7 to 9.2 (mean 4.1) mmHg. Mean gradient negatively correlated with an increase in the actual orifice area (ACA) derived from the valve orifice diameter given by the manufacturer (r = -0.45, P = 0.004). Mitral valve area calculated by both PHT and CE increased significantly with an increase in the ADA (r = 0.42, P = 0.007 and r = 0.32, P = 0.046, respectively). Mitrat valve area by the CE averaged 1.55 +/- 0.36 cm(2) (range 0.85 cm(2) for a 25 mm valve to 2.41 cm(2) for a 29 mm valve) and was smaller than by PHT (mean 2.04 +/- 0.41 cm(2), range 1.40-3.14 cm(2); P = 0.0001; t-test), irrespective of whether PHT is less than or >110 ms.Conclusion The Doppler parameters obtained with CHVP in the mitrat position are comparable with those obtained with the different prosthetic valves in common use. In the selected group of patients with CHVP, assessment of MVA by the PHT method is comparable with that by the CE. Areas by both methods were smaller than the AOA provided by the manufacturer, as seen in other similar design valves.|
|Education Level||UG and PG|
|Learning Resource Type||Article|
|Educational Framework||Medical Council of India (MCI)|
|Journal||EUROPEAN JOURNAL OF ECHOCARDIOGRAPHY|
Ministry of Human Resource Development (MHRD) under its National Mission on Education through Information and Communication Technology (NMEICT) has initiated the National Digital Library of India (NDLI) project to develop a framework of virtual repository of learning resources with a single-window search facility. Filtered and federated searching is employed to facilitate focused searching so that learners can find out the right resource with least effort and in minimum time. NDLI is designed to hold content of any language and provides interface support for leading vernacular languages, (currently Hindi, Bengali and several other languages are available). It is designed to provide support for all academic levels including researchers and life-long learners, all disciplines, all popular forms of access devices and differently-abled learners. It is being developed to help students to prepare for entrance and competitive examinations, to enable people to learn and prepare from best practices from all over the world and to facilitate researchers to perform inter-linked exploration from multiple sources. It is being developed at Indian Institute of Technology Kharagpur.
NDLI is a conglomeration of freely available or institutionally contributed or donated or publisher managed contents. Almost all these contents are hosted and accessed from respective sources. The responsibility for authenticity, relevance, completeness, accuracy, reliability and suitability of these contents rests with the respective organization and NDLI has no responsibility or liability for these. Every effort is made to keep the NDLI portal up and running smoothly unless there are some unavoidable technical issues.
Ministry of Human Resource Development (MHRD), through its National Mission on Education through Information and Communication Technology (NMEICT), has sponsored and funded the National Digital Library of India (NDLI) project.
For any issue or feedback, please write to firstname.lastname@example.org